Skip to main content
Wikispaces Classroom is now free, social, and easier than ever.
Try it today.
Pages and Files
list of experiments
Solubility book (3rd Edn)
To estimate the amount of solute dissolved in a saturated solution of 4-chlorobenzaldehyde in chloroform and methanol, by UV spectroscopy.
Solutions of known concentrations of 4-chlorobenzaldehyde was made-up in chloroform and methanol. UV spectra of further diluted solutions of each was obtained to determine the λ
(wavelength of maximum absorbance) of the aldehyde. Absorbance of solutions of different concentrations was also obtained.These solutions were made by diluting the previously prepared solutions. From the absorbance values, molar extinction coefficient (molar absorptivity)
ε, was calculated from Beer's law
determining the slope of a plot of absorbance Vs concentration. Using molar absorptivity and absorbance, concentration of the saturated solutions was determined. This was done by obtaining the absorbance of dilute solutions (known dilution) of the saturated solutions of 4-chlorobenzaldehyde in chloroform and methanol.
4-Chlorobenzaldehyde - CHCl3
212A-2 (0.01mM in CHCl3, 1cm pathlength)
Overlay CHCl3 solutions
of 213A-2, 213A-3, 213A-4, 213A-5 and 213A-6
Overlay CH3OH solutions
of 213A-2, 213B-3, 213B-4, 213B-5, 213B-6
The absorption maxima of 4-chlorbenzaldehyde in methanol was
reported to be 255 mu
ethanol it is reported
(p 135)to be and 254.4nm, with
ε of 16000
and 292nm with
bands, however it could not be found in chloroform. A plot of
Concentration Vs Absorbance
measured for 4-chlorobenzaldehyde in chloroform at 258.5nm wavelength, using 10mm path length gives an extinction coefficient of 14960
Absorbance obtained experimentally in methanol is strikingly different from observed in this experiment, 220nm. This may be due to the solvent/sample impurities or because of experimental uncertainty. Therefore this experiment will be performed again.
Absorbance of saturated solutions obtained are too low for an accurate measurement of solubility of the sample in the solvents used. Therefore this will also be repeated.
Extinction coefficient of 4-Chlorobenzaldehyde in Chloroform at 258.5nm has been determined to be 14960
. This is comparable to the reported values in methanol and ethanol.
13:22 Weighed out 4-chlorobenzaldehyde
(for solution 211A in CHCl3); 9.9959mM.
13:40 Weighed out 4-chlorobenzaldehyde
(for solution 211B in CH3OH); 9.9980mM
14:00 Made-up 100mL solutions of 4-chlorobenzaldehyde
(in Chloroform using 140.51mg) and
(in Methanol using 140.54mg).
five dilute solutions
using 212A, the new solutions are
213A-2, 213A-3, 213A-4, 213A-5
in chloroform in 10mL volumetric flasks. The concentration of each solution is listed in the spreadsheet
18:00 Obtained UV Wavelength scan for chloroform solution (213A dilute solutions)
19:00 Obtained Absorbance for the chloroform solutions (213A dilute solutions).
five dilute solutions
using 213B, the new solutions are
213A-2, 213B-3, 213B-4, 213B-5, 213B-6
in methanol in 10mL volumetric flasks. The concentration of each solution is listed in the spreadsheet
17:20 Obtained UV Wavelength scan for methanol solution (213B dilute solutions)
20:00 Obtained absorbance for the methanol solutions (213B dilute solutions).
1.Anything special about the procedure for doing the UV-Vis, how do I know the instrument is calibrated? Do you use a specific type of cuvette? What is the path length?-CN
Calibration check for the instrument had not been performed before the experiment, however it was done soon after and found to be accurate for all practical purposes. In the future a calibration report will be uploaded. Calibration or the Wavelength Accuracy check on the Perking Elmer lambda 35 UV-Vis was performed with a Deuterium Lamp Test,
peak at 656.1nm). The instrument passed this test.
for obtaining a UV spectrum on the
Perkin Elmer Lambda 35
instrument, includes Autozeroing the instrument in order to allow 100% Transmittance with no samples loaded. The instrument being used is a double beam instrument which is programed to subtract the absorbance of the blank (solvent) from the sample, this makes sure that the recorded absorbance is a result of the solute only. Since polystyrene cuvettes do not provide a window for the absorbance of interest (200-350nm) to be measured, quartz cuvettes (
) are used. The path length of these cuvettes is 10mm
help on how to format text
Turn off "Getting Started"